V 28 February 2013
Sizin üçün oyun:
Client Release Notes
v 1.28 - February 2013
Fix for temperature correction in CO2 calculations
Correct pH interval entry problem
v 1.27 - December 2012 - no general release
Various fixes to better and more simply implement
support of Seabird UIMM
Switch to use of Global driver to free up device 3 - phasing out driver
Hide control unless user requires it.
Fixes to various data display functions.
v 1.22 - September 2012
Fixed preferences error occurring on new/clean installs.
Client will not launch unit if f/w not up to date.
Enhanced error reporting slightly.
v. 1.21 - September 2012
Fixed Open Data File button so it only opens regular files, not hex.
Added 57.6K with suppression of ^type info to GlobalGB pull-down (PMEL Logging).
Added enable of f/w update to debugwin for cases where f/w update fails.
Tris offset now also encodes the mCP lot for pH units. (Left of decimal indicates mCP batch 0,1,2... etc.)
Temp offset added earlier but not noted in release notes.
Fixed problems with firmware update to make it much more robust.
New Temperature variable for pH units for corrected temp added - and columnsets updated to use it.
Fixed major problem w/ ImportHex - not reading configuration correctly - closed file before reading last string.
Disallow intervals less than 4 minutes for pH instruments and warn if less than 15 min/more than 3 hrs.
v. 1.20 - June 2012
Requires firmware 53
Firmware 53 fixes many minor bugs and adds support for Seabird UIMM
Firmware 53 fixes minor problem with blanks on CO2 units
Improvements to switching SAMI to 9600 - including
autoswitching of client
Client now reports what driver is running if connected to running SAMI
Improvements to pH calculations to account for changes in pumping performance
Change to pumping schedule of pH units that should increase deployment duration
Better support for SBE-37
via serial device menu
v. 1.10 - No general release - in-house test/dev
v. 1.06 October 19, 2011
Fixed problem with downloading from SAMI's with f/w < v.45
Users should update firmware after downloading. Cannot safely program older f/w units
Fixed occasional error in updating firmware on Windows
Added feature to make software update complain occasionally if it can't see internet
Fixed graphing so x-axis adjust works better
Fixed filename problem on 2nd
attempt at downloading
Changed year-day so if a new year occurs, it will go > 365 and not loop in graph
Added support for Aanderaa 3835/4835 long output (10 data fields)
Fixed duplicate resource problem - reducing app file size.
v. 1.05 August 22, 2011
Added ability to read in externally logged files via 'Import Hex File'
Faster serial port discovery on Windows machines (prefs window)
Improved error handling/reporting - user can e-mail error to Sunburst
v. 1.04 March 4, 2011
Firmware revision to 45 to support new memory on boards (rev E and later)
Fixes to 9600 comm
Fixes to serial comm in general to minimize
errors with boot loader
Improved reset command.
Adjusted display precision of outputs in data viewer.
v 1.03 (not released - changes carry to 1.04)
Added general serial driver w/ pull down menu. Support for Aanderaa 3835
Fixed t24 bug - where noon jumped to midnight in time selection
Fixed bug reporting wrong # extra pump cycles and sub-cycle delay
Added global property - Client version as string
Downloaded file now leads w/ date of d/l
and client version
Added ability to communicate at 9600 instead of 57.6K for long cables
Firmware to 34
Fixed default flush/reagent cycle values (CO2 only)
Added warnings for blank and sample intervals outside 'normal'.
Improved support for SB-37 CTD
Firmware revision to 31/32
Fixed f/w check so it doesn't cascade multiple windows on some Win platforms.
Corrected minor error in pH calculation (Jenny).
Fixed delay in opening on computers not connected to internet.
First non-beta release.
Created pH_ConstSal, which uses salinity constant for all pKa calcs in pH
Created pH_MeasSal which uses salinity measurements from Seabird 37 CTD
Blank field now updates with change of interval to 3.5 days
Added more tool tips.
Added support for Seabird 37SIP as Device 2. Polls device and stores output
Fixed firmware check so it doesn't come up mistakenly
Fixed mislabeled field in pH Calibration window (superuser only)
Incorporate firmware version 28
as part of pkg
Fixed 'preference creation' bug on XP. (Knock on wood.)
Investigated XP slow load, but found no solution.
Fixed 'pH flush' so it doesn't open valve (doh!)
Moved 'salinity default' control out of super user status
Refresh 'Read LED' button on change in SU status to make visible on PC
Allow super user to run very short pH cycles.
menu to select open windows
Adjusted some default settings for pH deployment
Added 'default salinity' field (superuser), set to 35
If SAMI-pH, no DI flush option in 'Pre-start'
Moved 'Calibration', 'Name SAMI' menu items to SuperUser only
Added tool tips to many controls
Moved 'Read LEDs' to SuperUser
Read/Edit text now in
SAMI menu and as push button
Added 'pH Flush' button to automatically run 50 DI cycles
Slightly improved the printing of graphs
Improved graph scaling - curves now hold only points zoomed to
Fixed pH calculation
- had old code from JN
Fixed 'About' screen so text is visible. (Cause: use of 'Ariel' font)
Fixed multiple firmware warnings on Windows.
Improved 'Read LED' screen output w/ labels
Added status of 'Parsing done' for Deployment text
Improved timekeeping references in config/data files
Start/Stop times are shown as GMT w/ local time in parens
GMT offset (e.g. -7) is shown next to local time.
Improved interface by disabling buttons when buttons' function not available
window sizing for Win platform
Fixed potential bug w/ preferences seen on one XP machine
Fixed driver selection for Superuser so verbose and average line up. Added back a pH driver
Fixed 'time out' radio button so it powers up RTS when selected.
Added serial record, and GPS record and handling of those records to return Long/Lat
changes to font
, spelling, etc.
Improved column set list editor.
Made version check occur on app start-up
Firmware req'd updated to 27
Added radio button option to not 'time-out' serial link and increased time-out to 3 minutes
Added 'Read LED's' function to allow blank stabilization check during calibration.
Added 'Open Data file' button
Added logo to front panel
Changed 'run blank first' to default for CO2
Default d/l filename now
includes SAMI name and date
Extensive improvements to graphing
Added second y axis
Added adjustment controls for all axes
Added primitive print function
Automatically selects some variables for display
Added internet 'check version' ability
Checks version # vs. that available on Sunburst website
Offers user opportunity to auto open browser to d/l page
Dostları ilə paylaş:
Verilənlər bazası müəlliflik hüququ ilə müdafiə olunur ©muhaz.org 2017